- Technical advance
- Open Access
- Open Peer Review
This article has Open Peer Review reports available.
A double-blind placebo needle for acupuncture research
© Takakura and Yajima; licensee BioMed Central Ltd. 2007
Received: 21 June 2007
Accepted: 10 October 2007
Published: 10 October 2007
Placebo needles that can mask acupuncture practitioners to the type of needle used have been considered almost impossible to develop until now.
We designed a double-blind non-penetrating placebo needle, the needle tip of which simply presses against the skin, and a matched penetrating needle. The needles are encased inside an opaque guide tube and the appearance and feel of the pair are designed to be indistinguishable. To validate the masking effect for the practitioner, 10 acupuncturists each applied 23 non-penetrating needles and 17 penetrating needles to the Large Intestine-4 point. After removing each needle, they judged whether the needle was 'penetrating', 'non-penetrating' or 'unidentifiable'. For the validation of patient masking, an acupuncturist randomly applied a non-penetrating/penetrating needle pair to the bilateral Sanjiao-5 points in 60 volunteers. When both applications were completed, we asked them to write down anything that they noticed regarding the needle application and associated sensations.
The mean ± SD of correct/unidentifiable/incorrect answers given by the 10 acupuncturists were 17.0 ± 4.1/6.4 ± 3.6/16.6 ± 3.0, respectively. Regarding patient masking, none of the subjects commented in the questionnaire that they had received a non-penetrating needle. Of 60 penetrating and 60 non-penetrating needle applications, 48 (80.0%) and 25 (41.7%) applications elicited skin penetration sensation and 48 (80.0%) and 20 (33.3%) applications elicited de qi, respectively.
These needles have the potential to mask both practitioners and patients from the type of needle used in acupuncture research.
The strongest evidence supporting the efficacy of acupuncture has been obtained using single-blind methods [1–4], which fail to meet the methodological standards for study blinding in conventional medicine [5–8]. As a result, the effectiveness of acupuncture has remained controversial, even though studies of the highest possible quality have been published in leading medical journals . The reason for this is that study subjects/patients are still exposed to possible bias due to the expectations, enthusiasm, suggestions and attitude of unmasked practitioners [5–13], and placebo needles aimed at masking practitioners have been considered unfeasible [7, 8, 13]. Until date, there have been no published reports on the development of a procedure or placebo needle that can mask practitioners from the type of needle used.
Although single-blind trials using placebo or sham needles are a significant advance [14–16], double-blind trials using placebo needles are critically important to ensure that acupuncture research meets the methodological standards of medical science to provide stronger evidence of the effectiveness of treatment using needles [5–8]. Only then will acupuncture be incorporated into generally accepted practice [5–8, 11–13].
Here, we report the design of double-blind (practitioner-patient masking) non-penetrating placebo and matched needles  to solve the methodological conundrum of practitioner masking [7, 8, 13, 14] with a statistical evaluation of the masking effect of these needles.
We recruited well-experienced and licensed acupuncturists on the teaching staff and healthy volunteers familiar with acupuncture treatment as experimental subjects from Hanada College. Before the study, the purpose and format were explained and the subjects provided written consent. The Showa University Ethics Committee gave its approval.
Design of double-blind needles
Validation test for practitioner masking
Numbers of correctly, unidentified and incorrectly identified needles in 10 acupuncturists on 40 (23 non-penetrating/17 penetrating) needles
Years of experience
Number of correctly identified needles (non-penetrating/penetrating)
Number of unidentified needles (non-penetrating/penetrating)
Number of incorrectly identified needles (non-penetrating/penetrating)
Mean ± SD
12.4 ± 7.8
17 ± 4.1 (10.7/6.3)
6.4 ± 3.6 (2.9/3.5)
16.6 ± 3.0 (9.4/7.2)
Validation test for patient masking
Sixty healthy volunteers (29.7 ± 7.5 years, 35 men, 25 women) who were familiar with receiving acupuncture were recruited for the validation study. Before the trial began, the experimental procedure was explained to the subjects as follows: 'We will apply two needles, which may or may not differ in type, at bilateral Sanjiao-5 (SJ-5) points that are located three finger widths above the wrist crease between the ulna and radius on the posterior surface of the forearm . Once both applications have been completed, we will ask you for each arm whether you felt a skin penetration sensation and de qi, a deep dull pain sensation that is considered essential for a successful acupuncture treatment . We will ask you to write down anything that you noticed, however trivial, regarding the needle application'.
Sixty penetrating/non-penetrating needle (10 mm insertion depth ) pairs were prepared and each needle was sealed in a small opaque container. The acupuncturist applied a pair of needles to each of the 60 subjects at the bilateral SJ-5 points, one needle in the right arm and the other in the left, using the alternating twirling technique. After each application, the subjects reported whether they felt a skin penetration sensation and de qi for each arm and wrote down anything that came to their notice. The practitioner guessed the authenticity of the needle after its removal. During this part of the study, the 'unidentifiable' option was discouraged.
The chi-squared test (SPSS version 15.0J; SPSS Inc, Chicago, IL) was used to determine whether the number of correctly and incorrectly identified needles fitted a probability of 0.5 and to compare the frequency of needle sensations between penetrating and non-penetrating needles.
Validation test for practitioner masking
The number of correct/unidentifiable/incorrect answers given by the 10 acupuncturists had a mean ± SD of 17.0 ± 4.1/6.4 ± 3.6/16.6 ± 3.0, respectively. Overall, the 170 correct and 166 incorrect identifications fitted a probability of 0.5 (χ2 = 0.048, p = 0.827), excluding the 64 unidentifiable needles. Furthermore, 107 correctly identified non-penetrating needles and 94 incorrectly identified non-penetrating needles (χ2 = 0.841, p = 0.359), and 63 correctly identified penetrating needles and 72 incorrectly identified penetrating needles (χ2 = 0.600, p = 0.439) fitted a probability of 0.5 (Table 1).
Validation test for patient masking
None of the subjects commented in the questionnaire that they had received a non-penetrating needle. Of the 60 penetrating and 60 non-penetrating needle applications, 48 (80.0%) and 25 (41.7%) applications elicited skin penetration sensations and 48 (80.0%) and 20 (33.3%) applications elicited de q i, respectively. The frequency of needle sensations was significantly different between non-penetrating needles and penetrating needles (skin penetration sensation χ2 = 18.502, p < 0.001; de q i, χ2 = 26.606, p < 0.001).
Of the 120 needles, the practitioners identified 65 (54.2%) correctly (penetrating needle = 35, non-penetrating needle = 30) and 55 (45.8%) incorrectly (penetrating needle = 25, non-penetrating needle = 30), which fits a probability of 0.5 (χ2 = 0.833, p = 0.361).
The practitioners failed to distinguish between the penetrating and non-penetrating needles, regardless of their practical experience. The subjects were also unable to distinguish needle authenticity.
For double-blind acupuncture studies, real and placebo needles must be identical for all variables except skin penetration. This is the ultimate aim underlying the design of the needles; they must fit these preconditions. The appearance and feel of the non-penetrating placebo and penetrating needles in this study were virtually identical, such that even well-experienced acupuncturists required deliberation to determine whether a needle was real or placebo. The findings that 16% of all needles were unidentifiable and that the practitioners identified approximately 50% of the other needles incorrectly indicate the potential of these needles in practitioner masking. Although our results suggest that skin penetration and further insertion of the needles were masked from the acupuncturists in this study, the needles must be validated in clinical settings with variables such as clinical improvement, adverse reactions and repeat treatments with multiple needles. These variables, as well as slight bleeding and patient reaction to strong pain elicited by real needle insertion in some instances, could break the blind.
Because our masking needle is designed for use at all acupuncture points, there should be no problem in its use on the toes, fingers or scalp, which are popular sites [7, 19]. It may be necessary to shave hairy skin sites to ensure firm adhesion.
Although skin penetration pain and de qi were less common with the non-penetrating needles than with the penetrating needles, none of the subjects suspected that they had received a non-penetrating placebo needle. This could be due in part to the fact that pain does not necessarily accompany needle insertion or removal, and that the subjects had previously experienced a very faint sensation elicited by the insertion of a fine needle. Indeed, approximately 20% of the penetrating needle applications in this study elicited neither skin penetration pain nor de qi. This suggests that the small proportion of study participants who did not sense needle penetration despite the insertion of a real needle might have guessed that a non-penetrating needle was being used if they had been informed of the possible use of the non-penetrating needles. Based on the findings of studies using single-blind needles [14–16], we believe that the subjects were unaware of the fact that we used non-penetrating needles due to the pressure applied to the skin by the blunt tip. However, the subjects were unlikely to report that they had received a non-penetrating needle, even if they suspected it, because we did not ask them whether they thought they had received a penetrating or non-penetrating needle. Thus, the successful subject masking observed in this study might indicate that our expectations were correct. The masking capability of the needle in study participants must be validated under conditions in which they have been informed of the possible use of non-penetrating needles, and subjects' speculations of the type of needle used must also be assessed . If needle sensations per se can induce physiological responses, the low frequency of needle sensations elicited by non-penetrating needles would limit their use as a placebo device.
The findings of the present study suggest that these double-blind placebo needles have potential for masking both acupuncture practitioners and patients. Given the recent developments in acupuncture research and practice, these needles should pave way for double-blind experiments to allow scientific assessment of the effect of acupuncture.
We thank Koji Sakurai (Hanada College, Tokyo) and Professor Ikuo Homma (Second Department of Physiology, Showa University School of Medicine, Tokyo) for their support. We also express our appreciation to all the participants of this study.
- Melchart D, Streng A, Hoppe A, Brinkhaus B, Witt C, Wagenpfeil S, Pfaffenrath V, Hammes M, Hummelsberger J, Irnich D, Weidenhammer W, Willich SN, Linde K: Acupuncture in patients with tension-type headache: randomized controlled trial. BMJ. 2005, 331: 376-382. 10.1136/bmj.38512.405440.8F.View ArticlePubMedPubMed CentralGoogle Scholar
- Berman BM, Lao L, Langenberg P, Lee WL, Gilpin AM: Effectiveness of acupuncture as adjunctive therapy in osteoarthritis of the knee: a randomized, controlled trial. Ann Intern Med. 2004, 141: 901-910.View ArticlePubMedGoogle Scholar
- Assefi NP, Sherman KJ, Jacobsen C, Goldberg J, Smith WR: A randomized clinical trial of acupuncture compared with sham acupuncture in fibromyalgia. Ann Intern Med. 2005, 143: 10-19.View ArticlePubMedGoogle Scholar
- Witt C, Brinkhaus B, Jena S, Linde K, Streng A, Wagenpfeil S, Hummelsberger J, Walther HU, Melchart D, Willich SN: Acupuncture in patients with osteoarthritis of the knee: a randomized trial. The Lancet. 2005, 366: 136-143. 10.1016/S0140-6736(05)66871-7.View ArticleGoogle Scholar
- Shapiro AK, Shapiro E: The powerful placebo: From ancient priest to modern physician. 1997, London: The Johns Hopkins University Press, -137-174, 190-216. 190–216Google Scholar
- Shapiro AK, Shapiro E: The placebo: Is it much ado about nothing?. The placebo effect: An interdisciplinary exploration. Edited by: Harrington A. 1997, Cambridge, Massachusetts, London, England, Harvard University Press, 12-36.Google Scholar
- Kaptchuk TJ: Placebo needle for acupuncture [Correspondence]. The Lancet. 1998, 352: 992-10.1016/S0140-6736(05)61551-6.View ArticleGoogle Scholar
- Kaptchuk TJ, Frank E, Shen J, Wenger N, Glaspy J, Hays RD, Albert PS, Choi C, Shekelle PG: Methodological issues in trials of acupuncture [Letter]. JAMA. 2001, 285: 1015-1016. 10.1001/jama.285.8.1015.PubMedGoogle Scholar
- Kaptchuk TJ: The placebo effect in alternative medicine: can the performance of a healing ritual have clinical significance?. Ann Intern Med. 2002, 136: 817-825.View ArticlePubMedGoogle Scholar
- Gracely RH, Dubener R, Deeter WR, Wolskee PJ: Clinicians' expectations influence placebo analgesia. The Lancet. 1985, 15: 43-10.1016/S0140-6736(85)90984-5.View ArticleGoogle Scholar
- Turner JA, Deyo RA, Losser JD, Korff MV, Fordyce WE: The importance of placebo effects in pain treatment and research. JAMA. 1994, 271: 1609-1614. 10.1001/jama.271.20.1609.View ArticlePubMedGoogle Scholar
- Kenneth FS, Iain C, Richard JH, Douglas GA: Empirical evidence of bias. Dimensions of methodological quality associated with estimates of treatment effects in controlled trials. JAMA. 1995, 273: 408-412. 10.1001/jama.273.5.408.View ArticleGoogle Scholar
- Kaptchuk TJ: Acupuncture: theory, efficacy, and practice. Ann Intern Med. 2002, 136: 374-383.View ArticlePubMedGoogle Scholar
- Streitberger K, Kleinhenz J: Introducing a placebo needle into acupuncture research. The Lancet. 1998, 352: 364-365. 10.1016/S0140-6736(97)10471-8.View ArticleGoogle Scholar
- Park J, White A, Stevinson C, Ernst E, James M: Validating a new non-penetrating sham acupuncture device: Two randomized controlled trials. Acupuncture in Medicine. 2002, 20: 168-174.View ArticlePubMedGoogle Scholar
- Fink M, Gutenbrunner C, Rollnik J, Karst M: Credibility of a newly designed placebo needle for clinical trials in acupuncture research. Forsch Komplementarmed Klass Naturheilkd. 2001, 8: 368-372. 10.1159/000057254.View ArticlePubMedGoogle Scholar
- Takakura N, Hanada College: Safe needle, placebo needle, and needle set for double blind. US patent. 6575992, 2003 June 10, PCT/JP00/03144. 2000 May, B1Google Scholar
- Pomeranz B, Berman B: Scientific basis of acupuncture. Basics of Acupuncture. Edited by: Stux G, Berman B, Pomeranz B. 2003, Berlin Heidelberg New York, Springer-Verlag, 146-148. 202, 51–55, 5Google Scholar
- Dincer F, Linde K: Sham interventions in randomized clinical trials of acupuncture-a review. Complementary therapies in medicine. 2003, 11 (4): 235-242. 10.1016/S0965-2299(03)00124-9.View ArticlePubMedGoogle Scholar
- White P, Lewith G, Hopwood V, Prescott P: The placebo needle, is it a valid and convincing placebo for use in acupuncture trials? A randomized, single-blind, cross-over pilot trial. Pain. 2003, 106: 401-409. 10.1016/j.pain.2003.08.013.View ArticlePubMedGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1472-6882/7/31/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.